3Remarks: Telescope run 527-531. Bias scan 100 V. Telescope got out of sync. Match with telescope data only up to 175000 events.
240 deg (parallel to tele. planes)
Remarks: Telescope run 535. Bias scan 100 V. Run crashed over night. Telescope sync only up to event 190000 (planned 2500000 events).
Remarks: Telescope run 546. Bias voltage 100 V. Beam position on only one chip. Run destroyed. Intended trigger count 500k
Remarks: Telescope run 551. Longrun over night. Run crashed. Events only up to 2288000 valid.
Remarks: Telescope run 579. Bias scan 10 V. run crashed. Data vaild up to 274915.
Telescope chiller turned off (could be the root cause of runs crashing).
First run for alibava modules with Sensor IFX_CB.
Beam energy = 5.6 GeV
Telescope threshold = 6
Telescope data file = 496
Bias voltage = -100V
Telescope synchornisation: yes
Position table (is center of sensor):
pulse shape looks good for latency 132. This value is now taken by default for all runs unless otherwise stated.
Beetle setting for a better defined pulse shape. These parameters were found by kazu, a beetle developer at lhcb.
Remarks: Beam position on only one chip.
Remarks: Telescope run 508. Bias scan 10 V.
Remarks: Telescope run 496. Bias voltage 100 V. First run at DESY TB.
Remarks: Telescope run 510. Bias scan 20 V.
Remarks: Telescope run 513. Bias scan 30 V.
Remarks: Telescope run 515. Bias scan 40 V.
Remarks: Telescope run 517. Bias scan 50 V.
Remarks: Telescope run 521. Bias scan 60 V.
Remarks: Telescope run 522. Bias scan 65 V.