Trying to get a nice plot of the beam profile by only using the PMTs 1 and 3 (horizontal) see http://elog.hephy.at/testbeam-SPS2015/66 for details.
Pos 2: x=69.3 mm, y=79 mm (Hits in APV 3-4 und 8-9 in p-Seite und 13 und 18 in n-Seite)
Producer.TLU Config Change:
Trying to get a nice plot of the beam profile by only using the vertical PMTs (number 0 and 2) see http://elog.hephy.at/testbeam-SPS2015/66 for details.
TLU DuT interface:
FIRRun002, PedestalRun007, Calibrationrun003
RUN WITH SVD3, telescope and FEI4
EUDET Run 64
Mimosa Threshold 6
Pos 5: x=30.5 mm, y= 99.74 mm (wide side top)
Room temp, fir002
EUDET Run 63
Pos 4: x=130.7 mm, y=84 mm
EUDET Run 62
Pos 3: x=38.6 mm, y=84 mm
cooled (see attachment) in -21.24 out -18.54
cooling supply emty at ~24:00 ie first 30k is cooled.
target: BW, pos 5
a working version of the EUDET online monitor (which works also with the FE-I4) is at the following location
~/BELLETEST/eudaq-neu/build/monitors/onlinemon$ ./OnlineMon.exe -sc 0 -f /home/telescope/eudaq/data/run000060.raw
EUDET Run 61
Apparently every problem can be attributed to a misalignment problem. The telescope has troubles with heavily misaligned DUTs. Was resolved by creating different gear-files for each run with pretty precise coordinates of the DUT position calculated for the telescope coordinate system.
Additionally one has to watch out for misaligned planes of the telescope itself. It was required to allow for more than 2mm offset for the telescope planes behind the DUTs. This can be easily checked by looking at the PreAligner plots in the -hitmaker-histo.root file.
It look something like this:
and not like this:
Hardware run with FEI4, Telescope and FW&BW Modules
Pos 10: x= 136.08mm, y=68.9 mm (narrow side down)
Bilder: Hitmap FE-I4 und Correlation zw. Mimosa und FE-I4
Testrun with the FEI4 working for the first time.
Lessons learned about the configuration:
EnableDUTVeto needs to be set for the FEI4 to allow it to work
HandShakeMode needs to be enabled for the FEI4 as well
"Enable RJ45 input during scans" needs to be enabled for the USB-controller
cooled (see attachment) in --21.24 out -18-54
target: fw, pos 9
RUN WITH SVD3 plus telescope
EUDET Run 58
120GeV hadrons, beam config H6A.BELLE.003
EUDET Run 57
Pos 9: x=30.5 mm, y= 60.67 mm (wide side down)
EUDET Run 56
Pos 8: x=136.08 mm, y= 79 mm (narrow side center)