Run to confirm trigger delay
0 offset (at the beginning)
Started APVDAQ and waited till 600 were taken then started EUDAQ
EUDAQ Run 21
Mimosa Thresholds lvl 6
Cooled, Temp see attachment
Telescope in combination with SVD3
pos x-Direction = Jura side
pos y-Direction = Down
Zero-point is in the top right corner in Beam direction
absolute Position of PI Stages:
Pos 1: x=100 mm, y=79 mm (Hits in APV 3 und 8-9 in p-Seite und 14 und 17 in n-Seite )
Pos 2: x=69.3 mm, y=79 mm (Hits in APV 3-4 und 8-9 in p-Seite und 13 und 18 in n-Seite)
Pos 3: x=38.6 mm, y=84 mm
Pos 4: x=130.7 mm, y=84 mm
Pos 5: x=30.5 mm, y= 99.74 mm (wide side top)
Pos 6: x=136.08 mm, y= 91.49 mm (narrow side top)
Pos 7: x= 30.5 mm, y= 79 mm (wide side center)
Pos 8: x=136.08 mm, y= 79 mm (narrow side center)
Pos 9: x=30.5 mm, y= 60.67 mm (wide side down)
Pos 10: x= 136.08mm, y=68.9 mm (narrow side down)
cooled (see attachment)
target: CE, pos 2
cooling box emergency break ... ie about the last 1-10 k not useable.
Something triggered a emergency break. HWRun012 only last 1-10k event has to be discarded.
120GeV hadrons, beam config H6A.BELLE.003
Testrun with the FEI4 working for the first time.
Lessons learned about the configuration:
EnableDUTVeto needs to be set for the FEI4 to allow it to work
HandShakeMode needs to be enabled for the FEI4 as well
"Enable RJ45 input during scans" needs to be enabled for the USB-controller
Hardware run with FEI4, Telescope and FW&BW Modules
Bilder: Hitmap FE-I4 und Correlation zw. Mimosa und FE-I4
Apparently every problem can be attributed to a misalignment problem. The telescope has troubles with heavily misaligned DUTs. Was resolved by creating different gear-files for each run with pretty precise coordinates of the DUT position calculated for the telescope coordinate system.
Additionally one has to watch out for misaligned planes of the telescope itself. It was required to allow for more than 2mm offset for the telescope planes behind the DUTs. This can be easily checked by looking at the PreAligner plots in the -hitmaker-histo.root file.
It look something like this:
and not like this:
RUN WITH SVD3, telescope and FEI4
EUDET Run 61
Mimosa Threshold 6
a working version of the EUDET online monitor (which works also with the FE-I4) is at the following location
~/BELLETEST/eudaq-neu/build/monitors/onlinemon$ ./OnlineMon.exe -sc 0 -f /home/telescope/eudaq/data/run000060.raw
EUDET Run 62
EUDET Run 63
EUDET Run 64
TLU DuT interface:
Trying to get a nice plot of the beam profile by only using the vertical PMTs (number 0 and 2) see http://elog.hephy.at/testbeam-SPS2015/66 for details.
Producer.TLU Config Change: