Testrun with the FEI4 working for the first time.
Lessons learned about the configuration:
EnableDUTVeto needs to be set for the FEI4 to allow it to work
HandShakeMode needs to be enabled for the FEI4 as well
"Enable RJ45 input during scans" needs to be enabled for the USB-controller
Apparently every problem can be attributed to a misalignment problem. The telescope has troubles with heavily misaligned DUTs. Was resolved by creating different gear-files for each run with pretty precise coordinates of the DUT position calculated for the telescope coordinate system.
Additionally one has to watch out for misaligned planes of the telescope itself. It was required to allow for more than 2mm offset for the telescope planes behind the DUTs. This can be easily checked by looking at the PreAligner plots in the -hitmaker-histo.root file.
It look something like this:
and not like this:
a working version of the EUDET online monitor (which works also with the FE-I4) is at the following location
~/BELLETEST/eudaq-neu/build/monitors/onlinemon$ ./OnlineMon.exe -sc 0 -f /home/telescope/eudaq/data/run000060.raw
Room temp, fir002
TLU DuT interface:
FIR filer and ADC delay with all sensor and telescope.