Cooled, Temp see attachment
Room temp, fir002
Room temp, fir004
temp: in -22 out -18,
cooled pedestal vaules for long sixlet cal
Information concerning the SVD3 setup during a beam test
In the beam area setup the whole crate and the power supply with the usual spider cable and the two Keithly 2410 high voltage supplies. Monitoring of the 2410 happens via a NI MAX interface that connects to the GPIB-Ethernet interface and needs to be configured there. The IP and password are noted on the device. The software can be found on heros in medialib/LONGTERM/longterm R3.1_doubleSided as a normal *.vi file.
Default adresses 16 and 17 as defined in labview, need to figure out on site which is which, if the voltage selected (e.g. 100V) is applied as +50/-50 or if it is applied as +100/-100 before connecting the cables.
Take a run without dut as well to check the telescope integrity. Without dut in beamline!!!!!
DHCP Range: 192.168.0.200 to 239
DNS Server: 192.168.0.254
external IP: 184.108.40.206 (für diverse Port Forwardings)
Manuell vergebene IP Adressen bitte unbedingt Benni mitteilen oder/und in der folgenden Tabelle eintragen!
x-Achse: positiv Richtung Saleve (counting room)
y-Achse: positiv Richtung UNTEN!!
position of laddertestbox xy stages:
Rev. Pos (BWD Sensor, BWD, Oben) x= 82.3, y=110.6
Pos 1: (mitte BWD sensor): x=144.74, y=80.8
Pos 2: (mitte Origami -z): x=269.62, y=80.8
Pos 3: (mitte Origami ce): x=394.42, y=80.8
Pos 4 (mitte FWD Micron Sensor): x=517, y=80.8
Pos 5 (Ecke BWD sensor): x=92.3, y=105.6
Pos 6 (Ecke Origami -z): x=217.18, y=105.6
Pos 7 (Ecke Origami ce): x=342.06, y=105.6
Pos 8 (Ecke FWD sensor): x=467,62, y=56,3
Pos 9 (Ecke FWD sensor): x=568,33 y=64,48
FIR filer and ADC delay with all sensor and telescope.
telescope n-side switched.
room temp max subevent at 2. with lat = 96
lat = 97 to a greater range of lat scan
lat = 98
Calibration Run with +- 50V Bias with higher compliance since the p-side draws more current than expected (23muA by the MAMBO)
room temp, FIRRun002, PedestalRun005, room temp
target: FW, pos 9 (p-0, n-128*4)
0 offset (at the beginning)
Started APVDAQ and waited till 600 were taken then started EUDAQ
EUDAQ Run 24
Mimosa Thresholds lvl 7
FIRRun002, pedestalrun006; temp see attachment in pedestalrun006
FIRRun002, PedestalRun006 (temp enty)
target: BW, pos 1
pos x-Direction = Jura side
pos y-Direction = Down
Zero-point is in the top right corner in Beam direction
absolute Position of PI Stages:
Pos 1: x=100 mm, y=79 mm (Hits in APV 3 und 8-9 in p-Seite und 14 und 17 in n-Seite )
Pos 2: x=69.3 mm, y=79 mm (Hits in APV 3-4 und 8-9 in p-Seite und 13 und 18 in n-Seite)
Pos 3: x=38.6 mm, y=84 mm
Pos 4: x=130.7 mm, y=84 mm
Pos 5: x=30.5 mm, y= 99.74 mm (wide side top)
Pos 6: x=136.08 mm, y= 91.49 mm (narrow side top)
Pos 7: x= 30.5 mm, y= 79 mm (wide side center)
Pos 8: x=136.08 mm, y= 79 mm (narrow side center)
Pos 9: x=30.5 mm, y= 60.67 mm (wide side down)
Pos 10: x= 136.08mm, y=68.9 mm (narrow side down)