Information concerning the SVD3 setup during a beam test
In the beam area setup the whole crate and the power supply with the usual spider cable and the two Keithly 2410 high voltage supplies. Monitoring of the 2410 happens via a NI MAX interface that connects to the GPIB-Ethernet interface and needs to be configured there. The IP and password are noted on the device. The software can be found on heros in medialib/LONGTERM/longterm R3.1_doubleSided as a normal *.vi file.
Default adresses 16 and 17 as defined in labview, need to figure out on site which is which, if the voltage selected (e.g. 100V) is applied as +50/-50 or if it is applied as +100/-100 before connecting the cables.
Take a run without dut as well to check the telescope integrity. Without dut in beamline!!!!!