Trying to get a nice plot of the beam profile by only using the PMTs 1 and 3 (horizontal) see http://elog.hephy.at/testbeam-SPS2015/66 for details.
Pos 2: x=69.3 mm, y=79 mm (Hits in APV 3-4 und 8-9 in p-Seite und 13 und 18 in n-Seite)
Producer.TLU Config Change:
Trying to get a nice plot of the beam profile by only using the vertical PMT 0 (upstream vertical, the long one) see http://elog.hephy.at/testbeam-SPS2015/66 for details.
RUN WITH SVD3, telescope and FEI4
EUDET Run 68
Mimosa Threshold 6
Pos 8: x=136.08 mm, y= 79 mm (narrow side center)
There are two planes, one fixed upstream and one fixed downstream.
CIS3WII_6 upstream and CIS2WI_IBL2 downstream.
They are configured using the pc in H6B which can be accessed via microsoft remote desktop from the network 192.168.5.x using the ip 192.168.5.6.
The configuration files can be found in the folder "E:\ConfigurationCMS\" on the server and are named after the modules.
Only one of the FEI4 detectors can be used at a time since they are of different generations (FEI4A and FEI4B) which can't be used at the same time with the available USB board. vnc
The configuration for EUDAQ needs to be adapted for the used module by changing the name of the configuration file.
SRAM_READOUT_AT = 5
SkipConfiguration = no
UseSingleBoardConfig = no
boards = 230
modules = 1
config_file = E:\ConfigurationCMS\CIS3_W11_06_thr1600_7to21ke_modified.cfg.root
#config_file = E:\ConfigurationCMS\CIS2_W02_IBL1_thr1600_7to14_modified.cfg.root
fpga_firmware = E:\icwiki_svn\USBPixI4\host\tags\release-5.3\config\usbpixi4.bit
lvl1_delay = 16
tlu_trigger_data_delay = 10
In my experience switching between the planes also requires a restart of the USBPix DAQ because apparently something is not working with the reload of a different config file afterwards. I had the phenomenon that the DAQ responded like it was working but the datastream was empty. Therefore always check the online monitor for hitmaps.
EUDET Run 71
Pos 1: x=100 mm, y=79 mm (Hits in APV 3 und 8-9 in p-Seite und 14 und 17 in n-Seite )
Downstream FEI4 used
EUDET Run 73
Pos 7: x= 30.5 mm, y= 79 mm (wide side center)
Upstream FEI4 used, checking if alignment change helped
EUDET Run 74
Upstream FEI4 used