Hardware Run +-50V Bias
Pedestal 018 (again marked all the bad strips)
Geometry setup of FW/BW:
(all dimensions are in mm and refer to the sensor planes)
Hardware Run +-60V Bias
Excellent, brilliant run.
cooled see appendix (FirRun003)
multi 6 Raw Mode
FIFO size increased
L6 are swapped, needs to be corrected later in the offline analysis.
cooled at -25
latest ADC delay and FIRRun
100k trigger dead time
Need to change L6n apv definition to:
mod = 9,0,1,1,1,m,29,0,0,0,0,0,0,L6_n_side
apv = 9,1,34,3,30,x,0,0,5,1,7,0,0,x
apv = 9,0,36,3,50,x,0,0,5,1,6,0,0,x
apv = 9,3,38,3,30,x,0,0,5,1,9,0,0,x
apv = 9,2,40,3,30,x,0,0,5,1,8,0,0,x
to get the right mapping of the sensor.
Initial pedestal run with SVD3 setup and WedgeOld, FW993, BW993 (reading out 4 chips on every p-side)
Initial calibration run with SVD3 setup and WedgeOld, FW993, BW993 (reading out 4 chips on every p-side)
Hardware run, hitting APVs 0/1 on p-sides, 1/2 on n-sides
EUDAT telescope running with difference of 2 events
Hardware run, hitting APVs 1/2 on p-sides, 1/2 on n-sides
Hardware run, hitting APVs 2/3 on p-sides, 1/2 on n-sides
cooled (see last calibration run)
Latest noise, FIRfilter settings.
Failed due to VME read timeout.
cooled see attachment in FIRRun_003
cooling see attachment of FIRRun003
Multi 6 raw mode
L5_n_side out of order
FIFO_FULL_ERROR starting at ~2000 events
same as hardware run 009
FIFO FULL ERROR starting at ~event 5000
FIFO FULL at event ~27000
Cooled down to -20
Bias voltage set to +-80V