Windows PC at Telescope:
0.) JTAG Programming: Chose a threshold file (recommended threshold 5) File: M26TB_Cern_2011.thresh5.mcf--> press start
1.) C:\Users\telescope\Desktop\users\BelleII --> start Mimosa 26 Telescope --> select all six Mimosa's
2.) C:\Users\telescope\Desktop\users\BelleII --> start startEudaq_fullWindows.bat
3.) Use configuration file hephy1 (location: C:\eudaq\conf). Change it according to your TLU settings (DUT mask, AND mask, etc.)
x) Online monitor in preparation
Files: Are saved in C:\eudaq\data with internal numbering "run000000xxx.raw" (look at the date and time). [You can connect locally to this folder at HEPHY-SVD PC]
Temperature: See attachment
Layer 3-n side deactivated, because Hao broke it.
See latest Noise, and PedestalRun.
Infineon new stack - batch 4
cooled (temperature see attachment latest noise run),
latest Noise, FIR settings with the right layout mapping.
cooled (see last calibration run)
Latest noise, FIRfilter settings.
Failed due to VME read timeout.
cooled, latest noise, cal, PedestalRun.
Same as HardwareRun_003 but with ZS.
cooled with new delay run
temp see attachment (von flo)
latest fir filter (002)
temp cooled see attachment
last noise run(007)
new cal run Raw, test with put fifo full flags
Cooled see appendix
with new layout (l3_n-side channels changed and reactivated).
cooled see appendix (FIRRun003)
cooled see appendix (FIRRun_003)
cooled see attachment in FIRRun_003
cooling see attachment of FIRRun003
Multi 6 raw mode
L5_n_side out of order
FIFO_FULL_ERROR starting at ~2000 events