At room temperature
Latest delay and FIRRun
Operating with AIDA telescope: TLUControl.exe.exe -d 3 -a 11 - i RJ45
Dut Position (left to right): 11, 06, 05, 07
First run was at: 23 (no track)
Trigger delay: 26
Trigger delay: 29
Trigger delay: 32
Trigger delay: 20
Trigger delay: 17
Trigger delay: 14
Trigger chip delay: 110
Latest ADC scan and FIR run.
Position: 0.0, see elog entry.
We set the coordinates trying to put the beam in an area with well working APVs
Peak of course differs for n and p sensors due to their different pitch.
L4-6 n-side Peak around strip 180
L4&6 p-side mean value around strip 380
L5 p-side moved by 4.8mm due to reasons (APV misalignment I think) therefore the peak of the same measurements is moved by 4.8/75µm = 64 strips therefore centered around strip 320
The coordinates of the x-y table were set to 0:0 at this point. Refer to Run Alignment2 for reference data.
Windows PC at Telescope:
0.) JTAG Programming: Chose a threshold file (recommended threshold 5) File: M26TB_Cern_2011.thresh5.mcf--> press start
1.) C:\Users\telescope\Desktop\users\BelleII --> start Mimosa 26 Telescope --> select all six Mimosa's
2.) C:\Users\telescope\Desktop\users\BelleII --> start startEudaq_fullWindows.bat
3.) Use configuration file hephy1 (location: C:\eudaq\conf). Change it according to your TLU settings (DUT mask, AND mask, etc.)
x) Online monitor in preparation
Files: Are saved in C:\eudaq\data with internal numbering "run000000xxx.raw" (look at the date and time). [You can connect locally to this folder at HEPHY-SVD PC]
Temperature: See attachment
Layer 3-n side deactivated, because Hao broke it.
See latest Noise, and PedestalRun.
cooled (temperature see attachment latest noise run),
latest Noise, FIR settings with the right layout mapping.
cooled (see last calibration run)
Latest noise, FIRfilter settings.
Failed due to VME read timeout.
cooled, latest noise, cal, PedestalRun.
Same as HardwareRun_003 but with ZS.
cooled with new delay run
temp see attachment (von flo)
latest fir filter (002)
temp cooled see attachment
last noise run(007)
new cal run Raw, test with put fifo full flags
cooled see appendix (FIRRun003)
cooled see appendix (FIRRun_003)
1.) Folder EUDAQ OFFLINE Monitor
1.) open command line and chose the path where OfflineMonitor is
2.)Type -i "patch of raw file" -o test.root 100
3.) helpfile typing -h