Preliminary run list for pt-modules + ite512 + silc1, silc2, silc5, silc6
playing with TRIM4 in vertical plane (setting to 20A current) shifted beamspot back to center position
Beam configuration file: H6b.802 (FM HAD (P0 off, H8=+180) PARLALLEL IN H6B SILC 2008
Rate: 3.35E5 particles/spill (measured with SCINT8 in beamline)
Particle Production Applet
For +120 Gev/c beam:
Data of the run was dicarded.
Compressed air flow was at ~2 bars during run. Graph shows RH rise after switching of CA flow. Restarted at around 7% with very low pressure.
Timescales for RH and I graph are different!!!
after 3 hours first bottle of compressed air (dewpoint -170 degree C) is gone and sensor Pt-h01 starts to fluctuate again
HV=70V, Current: 25uA first, 29uA then, 35uA latest. now stable
Wolfgang - Building: 41, Room: 204, Tel: ?
Manfred: Building 38, Room 522, Tel: 70522
Chrisu: B39, room 309, Tel: 79309
Bergi: Tel Zimmer: 79112, Handy: 16-2471
Maks: Tel Zimmer: 79117
manually changed TRIM3, TRIM4 and TRIM6
Kendall tau (rank correlation) for module 1 (single layer 1 / h13), module 2 (single layer 2 / h14), module 6 (standard sensor 2 / h12)