CO2 cooling: P_TT_4: 18.2 bar, mf=1 g/s
beam on; xy table position x=306 y=76.5
I=19.5µA; old I2C settings; corrected trigger delay (htr); adjusted front-end voltages
Temp. at pipe inlet (K7) = -14°C
Temp. at pipe outlet (K8) = -17°C
Before this run, the front-end voltages (+2.5 and +1.25) were readjusted so that the nominal values appear on MAMBO. (Previously, both were about 50mV too low.) Not sure if this makes a big difference, let's see.
06.10.2011 15:54:24 event=2901 rate=36events/s temp/curr=35.5/-05.6 35.5/-06.0 35.4/-06.9 34.9/001.9 35.0/002.2 35.0/019.6
06.10.2011 16:24:25 event=81350 rate=50events/s temp/curr=34.9/-05.6 34.9/-06.1 34.9/-06.9 34.3/001.7 34.3/002.2 34.3/019.6